In this datasheet you can get info like differential pressure, ambient temperature, installation, and transmitters. The datasheet presents things related to calibration, range, signal, and sensor.
Lots of information such as transmitter output, pressure transmitter, transmitter, and pressure are explained inside this datasheet. Here are some excerpt from this datasheet:
Two options are available as Output % Reranging operation: 1) OP Range Low where both LRV and URV are adjusted 2) OP Range High where only URV is change in accordance with the new input percentage As example: Actual level measured by the transmitter: Transmitter output = 27% Calibration : LRV = -125 mbar URV = +340 mbar a) New input level measurement (Option 1) = 30% New calibration : LRV = -139.5 mbar URV = +325.5 mbar The transmitter output is now = 30% Starting again from the initial settings: Transmitter output = 27% Calibration : LRV = -125 mbar URV = +340 mbar + – 44 – b) New input level measurement (Option 2) = 30% New calibration : LRV = -125 mbar URV = +291.5 mbar The transmitter output is now = 30% . a) Certificate ATEX II 1G, EEx ia IIC T5 (-40°C ≤ Ta ≤+40°C) respectively, EEx ia IIC T4 (-40°C ≤ Ta ≤+85°C) TUV Certificate number TPS 04 ATEX 1 008 The meaning of ATEX code is as follows: II : Group for surface areas (not mines) 1: Category G: Gas (dangerous
In addition, the datasheet presents discussion such as application, configuration, measurement, and output.