Inside the technical information you can find info like measuring range, electrode plastic gel, sensor, electrode, long connecting cable, measurement, waste disposal, system, temperature probe, electrode amplifier, handling, and probe. This technical information presents information related to connector, thermocouple, measurement procedure, instrument, measurement range, power supply, interface, port, range, operation, connection, and housing.
There are lots of explanation related to switch, force sensor, amplifier, surface, operating instructions, start measurement, measured value, carry out measurements, unit, socket, technical specifications, and display are presented inside this technical information. Below are grabbed from the technical information:
The Force Sensor serves for the computer-supported measurement of forces. It can be used in place of a springtype force measuring device in many experiments. A slide switch on the front of the housing allows switching between the two measuring ranges. When the slide switch is to the left, the ±10 Newton measuring range is activated, when it is to the right, then the ±50 Newton measuring range is activated. The hook that is screwed into the under end of the housing can be fixed in the particular position required with a plastic locknut. It allows the Force Sensor to be loaded with tensile or compressive forces. The Force Sensor can be held in experimental set-ups by inserting a stand rod through the borehole in the upper end of the housing. This borehole is 1 fitted with a milled screw with which the Force Sensor can be fixed in position.
In addition, this technical information tells the reader discussion things like sensor housing, force, cable, instructions, connecting cable, electrode plastic, pin, voltage probe, characteristics, interface software, and oxygen probe.