Many explanation related to resolution, output, and analog output are described in this installation instruction. The installation instruction presents info around sensor, and laser sensor.
In this installation instruction the reader can find things such as laser, and measuring range. The following are selected from this installation instruction:
Current Consumption Circuitry Protection Features • • • • • • • • • Output Type Visible red Class 2 laser 20 μm resolution 40 mm measuring range 0…10V DC analog output IP 67 enclosure 270° rotatable connector No user adjustments Contamination indicator Self-contained sensor Output Rating Response Time Housing Material Lens Material LED Indicators Connection Type Visible red Class 2 laser, 670 nm <0.8 mm beam spot @ 65 mm 45…85 mm 40 mm <1% 20 µ m 18 µ m/°C 18…28V DC ≤35 mA @ 24V DC Short circuit, overload, false pulse, transient noise, reverse polarity protection Analog Output 0…10V DC 3 mA max.
Furthermore, the installation instruction gives you info regarding measurement, and range.