The technical information gives us info related to code, applications, measurement, extract, pressure, power, pressure measurement, and tube. Lots of things around static pressure, information, transducer, output, controls, linearity, and range are explained inside this technical information.
In this technical information the reader can learn things related to pressure range, measured units, signal, display, sensors, digit, traceable calibration certificate, and calibration certificate. The following are grabbed from this technical information:
The P-Sensor’s electronic is easily accessible by removing the lid. Connect a Voltmeter either to the Terminals 1 and 2 or use a molex connector on the LCD Display connector J7. The output signal on the LCD connector is always 0…10V whilst the output on Terminal 1 and 2 depends on the Red Jumper setting. If set to the left position the output is 4…20mA. If set to the right position the output is 0-10V. It is important to know that the P-Sensor’s base calibration is based on 0…10V, which means, all calibrations must be carried out in 0…10V first.
Furthermore, this technical information gives us discussion about output signal, room pressure, supply, selection, sensor, handling unit, and calibration.