Lots of info about signal, power signal, power interface, power supply, distance, and the measurement point are described in the user manual. Inside this user manual you can learn information like measurement range, parameter, laser, configuration, and interface.
This user manual gives us things about configuration settings, voltage output, accuracy, the measurement range, and outputs. The following are some excerpt from this user manual:
Command Name Length Command Code Default Setting Set Zero-Point / Limit 1 1-6 bytes ASCII Code: Z[
] (0 ≤ ZeroPoint ≤ 50000) Function button code:1 Zero range (Z0) Set Limit Directions 2 bytes ASCII Code: F[ ] (1 ≤ LimitCode ≤ 4) Function button code:None F1 Active Inward F2 Active Outward (F1F4) Set Limit 1 1-6 bytes ASCII Code: J[ ] (1 ≤ LimitPoint ≤ 50000) Input Switch code:None Min range (J1) Set Limit 2 1-6 bytes ASCII Code: K[ ] (1 ≤ LimitPoint ≤ 50000) Input Switch code:None Max range (K50000) Set Span / Limit 2 1-6 bytes ASCII code: U (0 ≤ Span ≤ 50000) Function button code:2 50000 (U50000) Set Sample Interval (in 100μs increments) 2-6 bytes ASCII Code: S (8 ≤ Interval ≤ 50000) Function button code: 3 5 samples/second (S2000) Analog Output Mode 2 bytes ASCII Code: X (Mode:1 = Current, 2 = Voltage, 3 = NPN, 4 = PNP, 5 = off) Function button code: 4
Additionally, the user manual presents more things like the serial port, measurement, default configuration, range, and the power signal.