This datasheet tells you explanation related to the measurement range, distance, measurement, beam, response, and linear output. Many discussion regarding laser beam, inclination angle, sensitivity, sensor, displacement sensor, and offset adjustment are explained inside this datasheet.
Inside the datasheet the reader can read information regarding measurement range, inclination, laser displacement sensor, displacement, current consumption, and laser emission. Here are excerpted from this datasheet:
Laser OFF Input The laser-OFF input controls laser emission. Laser emission is turned ON when there is no laser-OFF (Amplifier) input (with a current leakage of 0.1 mA max.) and it is turned OFF when the laser-OFF input is short-circuited to the 0-V terminal (with a residual voltage of 2 V max.). When laser emission is turned OFF, the linear output, indicators, and the enable output of the sensor are on hold. There will be a drift of 0.1% FS/s when the analog output of the sensor is on hold. The response time required to turn laser emission ON or OFF is 3 ms max. Range The FAR green indicator and NEAR green indicator will be lit when the target is within the Indicators — measurement range of the Sensor.
Even more, the datasheet gives the reader information around amplifier, error, mounting, resolution, and laser.